Precise Testing Ultrasonic Horn Analyzer
For ultrasonic components measuring instrument, the most important is the admittance chart, logarithmic graph of the two graphics (the two graphics is the main method of qualified qualified), and more than a dozen parameters: resonance frequency Fs, the maximum conductance Gmax, bandwidth F2-F1, anti-resonance frequency Fp, quality factor Qm, free capacitance CT, dynamic resistance R1, dynamic inductance L1, dynamic capacitance C1, static capacitance C0, effective electromechanical coupling coefficient Keff.
|Product Feature||portable,8inch screen,full touch screen|
|Measurement index||All the parameters ,the graph|
|Measurement Accuracy||< 0.5%|
|Measuremet Speed||5/S one pass (600scan point) adjustable|
Parameters and Graph:
* Reduces ultrasonic equipment maintenance costs because it allows preventive maintenance
* Allows the quality control of ultrasonic transducers reducing technical assistance
* Makes fast and accurate manufacturing of transducer possible
* Easy operation
* Can connect to PC operation for storage
Q:What is Fs and Fp means?
A:Fs means resonant frequency, Fp means anti-resonant frequency.
Q:What is C 0,C T,C 1 means ?
A:C o means static capacitance, C T means free capacitance,C 1 means dynamic capacity.
C T= C 1+C 0
Q: Is quality factor Qm the higher the better?
A:Not exactly right.At a certain range, the Qm higher,the ultrasonic efficiency and sensitivity will also higher.But if the Qm is too high, the transducer can not match the generator and will not work effective.