Measuring Instrument Trz Horn Analyzing
The TRZ Horn Analyzer determines the operation frequency of the item to be tested and its impedance and current with the touch of a button. It’s quick and easy. With this quick and easily retrieved information, the user can perform the quality control of transducers, tuning of sonotrodes and ultrasonic horns, preventive and corrective maintenance of acoustic sets and much more.
|Product Feature||portable,8inch screen,full touch screen|
|Measurement index||All the parameters ,the graph|
|Measurement Accuracy||< 0.5%|
|Measuremet Speed||5/S one pass (600scan point) adjustable|
Parameters and Graph:
1. Large screen, full-screen touching, making the HS520A both HS70A-P(computer version) operating experience, and no cumbersome with a sense of computer operation.
2. Relies on the new powerful ARM processor data processing capacity, HS520A measurement accuracy is much higher than HS70A-D (with display version), can achieve HS70A-P(computer version) level.
3．HS520A was taken into account the needs of automated production line operation in the development time, reserved the 232interface with PLC communication and IO communication interface (replay input and output)
4. HS520A operation is no longer based on Windows system, with long-term reliable operation capacity in the production line.
Q:How to judge the defects for the ceramic on you analyzer?
A:For piezoelectric ceramics, the merits of the device can be judged directly from admittance circle plots and logarithmic coordinates. If there are delaminations or cracks inside the ceramic plate, the logarithmic curve will appear multi-peaked, and the admittance circle diagram will appear more than one Parasitic small round.
Q:How to judge the defects for booster and assemble?
A:For the design, processing and assembly of the horn or booster , whether it is reasonable or flawed, it can be clearly seen directly on the guide circle diagram.
Q:what kinds of graph will display on your impedance analyzer?
A:There are G B-F; LgZ Phi-F; R`X-F; Z Phi-F; Y Phi-F graphs.